Operator did not ensured the Inner Diameter of the part to the drawing requirement. |The operator did not calculated the Air gauge reading properly.|The Air gauge was set to Dia. 19.05mm and it has to be measured with in +21Micron, but the operator calculated the air gauge reading as 19.00mm and +25Micron.|Inadequate knowledge on the measuring instrument.